Replacing Probe Extractor Seals
The extractor assembly* probe seal uses a set of interlocking packing seals and a compression nut that should only require hand tightening. If a leak is observed around the probe, further hand tightening may be required, however over tightening of the compression nut using wrenches may damage the seals. If the leak cannot be stopped, the packing seals should be replaced. The following procedure illustrates how to replace the packing seals. Replacement seals are PN 45096-00001. *When inserting the probe with the extractor assembly always be careful not to insert too deeply, causing damage to the probe tip. A collar type “stop” may be added to the probe or lead screws to prevent over-insertion.
Terms and Conditions of Sale - Extrel
Sellers-Agreement-Purchasing-Terms-PUR-0026-FR-rev3 - Extrel
Terms and Conditions – Process Insights Swiss AG - MBW
Terms and Conditions – Tiger Optics
Terms and Conditions – Process Insights AG - LAR
Terms and Conditions – Guided Wave
Terms and Conditions of Sale – COSA Xentaur
Terms and Conditions – Alpha Omega Instruments
Copyright © 2023 Process Insights, Inc. All Rights Reserved.